How do I know what size of ArcAngel to use?
Sizing of the Arc is based on the size of the ET tube size used.
* 5F inline suction catheter is recommended.
** Will secure 6F & 8F EDI catheters.
How are tubes adjusted independently of each other?
The innovative twist-lock technology allows for keeping one tube secure while the other can be freely adjusted. Open the locking ring to the desired tube holder groove. Make your adjustment by advancing or retracting your tube while remaining in the groove. Twist ring to lock in place.
How many people does it require to make an adjustment to a secured tube? Is a tape measure required?
It takes one clinician seconds for the adjustments. No tape measure is need as you can use the 0.25cm hashmarks on the outer ring as your guide.
How do I remove the ArcAngel in an emergency?
In an emergency or for trial extubation, the ArcAngel can be removed by holding the cheek pads in place, gently pulling the ArcAngel out of the cheek pads, leaving the pads and any additional tube intact. The same ArcAngel can be inserted back into the cheek pads for continued use if needed.
Can you use in-line suction for the ETT with ArcAngel?
Yes. There is no outer compression on the tube. It is held in place by friction alone.
Does ArcAngel contain latex or metal?
No. It is MRI safe.
Where do I place the cheek pads?
Simply place the cheek pads directly in front of the ear. Nasal vs oral intubation will determine the position height to insure the proper alignment of the ETT. Designed with baby’s facial contour in mind there is no need to trim the cheek pads. There are 2 interchangeable sizes available for a custom fit.
*Refer to Directions for Use.
How do the cheek pads adhere to the face?
An oversized adhesive is used to secure the cheek pads. In clinical trials, adhesive has remained secure for 12 days with 80% humidity.
What happens if the cheek pads get soiled?
They can be cleaned with a baby wipe or alcohol swab while remaining on infant.
How are the cheek pads removed?
Wipe the external surface of adhesive generously with Nurse Angele’s Wipes or your adhesive remover of choice, allowing time for full saturation of the adhesive. Next, gently pull up a corner of the pad and continually wipe the skin/adhesive interface while gently pulling off the cheek pad.
Why are there 3 grooves on the top and 3 grooves on the bottom?
Any tube (ETT, feeding, or gastric suction) being inserted nasally will use the top 3 grooves. Any tube inserted orally will use the bottom 3 grooves.
Can the ArcAngel be reused?
The ArcAngel can be used for the entirety of the patient’s stay if the ETT remains the same size. Cheek pads can be replaced if needed. It is single-patient use.
What if my patient needs one size ArcAngel but a different size tube won’t fit in the grooves?
The ArcAngel can be used alongside additional adhesive if needed.
Redefining the Lifesaving Tools that Protect Our Most Vulnerable.
With over 20 years of first-hand experience, WarriorNP has a vision to equip and empower every NICU and PICU worldwide with innovative patient-first medical devices and products.Contact